• For hot sanitary water, drinkable water• pH must be constant (for free chlorine)• Temperature: up to 50 °C or up to 70 °C• Pressure: up to 8 bar
Concentration measurement of free chlorine or chlorine dioxide, hot sanitary waterand drinkable water.Caution: The pH must be constant for free chlorine measurement.
Disinfection additives: Inorganic compounds, such as NaOCl, Ca(OCl)₂, gaseous chlorine, chlorine produced by electrolysisChlorine dioxide: Chlorite/ Chlorine; Chlorite/ AcidMeasuring system: Semi-open cell with 3 electrodes and electrolyteOutput signal: 4 ... 20 mAOperating temperature:
Automatic temperature compensationAmbient temperature: From 0 to 55 °COperating pressure: 8 bar max.Cleaning system Equipment: RV1 (option)Flow rate: Flow must be constant
pH operating range:
Zero adjustment Not necessary (from factory)Slope calibration: Only 1 point necessary with BAMOPHAR 194: